NHG60 Touchscreen GlossMeter for marple,granite,polishing
brick,ceramic,and mold indust
NHG60 gloss meter features with 60° angle, manufactured according
to ISO2813 and GB/T 9754. It is also compatible with the standards
of ASTM D523, ASTM D2457. NHG60 conforms to JJG696 first class
gloss meter working requirement, corresponding with CIE 1931(2°)
under CIE C light source.
NHG60 gloss meter can test material with gloss (0-1000Gu), and
universally apply to paint, ink, stoving varnish, coating, wood
products; marble, granite, vitrified polished tile, pottery brick
and porcelain; plastic, paper; hardware industries, etc. for gloss
measurement and gloss data transmission.
1. 60° angle for semi gloss
2. Display 5 sets of measurement data, good for comparison
3. Measuring mode: basic measurement, statistical measurement,
continuous measurement, QC measurement (only for GQC6 software).
4. Continuous mode to check uniformity over large areas
5. Connect to PC, more extend functions
6. Input gloss value manually
7. Large touch screen operation, easy to use
8. Beauty appearance, good man-machine communication interface
9. Powerful functions to meet many different requirements
10. Large storage to save over 5000 data
11. Built-in lithium ion rechargeable battery.
NHG60 60° Precise Gloss Meter
Comply with standard ISO 2813,GB/T 9754,ASTM D 523,ASTM D 2457
Measuring Area (mm)
Measurement Range Repeatability Reproducibility
0-10GU 10-100GU 100-1000GU
± 0.1GU ± 0.2GU ± 0.2%GU
± 0.2GU ± 0.5GU ± 0.5%GU
Conform with JJG696 first class gloss meter working requirement
Corresponding with CIE 1931(2°) under CIE C light source
±1.2 , ± 1.2%
L*W*H : 160mm*75mm*90mm
3200mAh Li-ion Battery, >10000 times(within 8 hours)
TFT 3.5 inch, capacitive touch screen
Basic mode: 1000, Statistical Mode: 5000, Continuous mode : 5000
GQC6 Quality Control Software with QC report printing functions and
more extended functions.
<85% relative humidity, no condensation
Power Adapter, USB cable, User Manual, CD (Including QC software),
The specifications are subject to change without notice.
Keywords: gloss meter, glossness meter, glossmeter